Think & Tinker, Ltd.
P.O. Box 1606, Palmer Lake, CO 80133
Tel: (719) 488-9640, Fax: (866) 453-8473
Sales: Sales@thinktink.com, Support: Support@thinktink.com
If you have not done so, please read the "DOs and DON'Ts section now
- Titration flask (125 ml)
- 10 ml Automatic pipette
- Digital titrator
- Deionized water
- Phenolphthalien indicator
- 1.60N Sodium hydroxide (titrant)
- Measure 30 ml of deionized water into the titration flask. The actual quantity of water is not critical, but too much will make the end point too faint and may result in an erroneous reading.
- Using the automatic pipettor, add 1.0 ml of sample solution.
- Add 1 to 4 drops of phenolphthalien indicator to the flask. Swirl the flask around to mix thoroughly. The solution will be clear after the addition of the indicator (if there is acid present).
- Using the digital titrator, titrate with 1.60N sodium hydroxide until the solution turns red. This is a fairly subtle endpoint, so watch closely.
- Assuming that your bath is not excessively depleted of acid, the digital titrator should read at least 600 before you can expect a color change. A bath with the optimum amount of acid present will yield a reading of 821 (± 10).
- Record the number of drops of titrant used as indicated on the digital readout.
- The amount of SolderOn acid that is needed to replenish the 38 liter (10 gallon) bath can be calculated using the relationship (the digital titrator reading is denoted NUMACID):
|SolderOn Acid needed (liters) = (821 - NUMACID) x 0.0106 liters
In imperial units (US gallons), the amount of acid needed is expressed as:
|SolderOn Acid needed = (821 - NUMACID) x 0.0028 gal.
To express the concentration of SolderOn acid as a percent of bath volume, use:
|% SolderOn acid = 0.0280 x NUMACID
The acceptable range of digital titrator readings is:
|Acceptable Range: percent:
||20 to 35%
||714 to 1,250
Always use the proper attire as outlined above in the SAFETY section when handling concentrated sulfuric acid. If contact should occur, flush affected area with cool tap water for 15 minutes. CONTACT A PHYSICIAN IMMEDIATELY.
HANDLE WITH CARE
Always wear protective clothing, acid resistant gloves, and a full coverage face shield when handling this or any other corrosive material.
On the web since 1994
Sales: 1-(719) 488-9640 Tech Support: 1-(719) 488-9640 Fax: 1-(866) 453-8473
Copyright © 1994 - 2014 Think & Tinker, Ltd.